Question about ESCHMANN Little Sister Autoclave Tabletop
The autoclave goes through the entire cycle ending with "drying" but the water doesn't drain (it pours out when opening the autoclave). Does anyone perhaps know what the problem is and how to fix it?
Posted by Anonymous on
Check the drain pipe that vents steam through resevoir at end of cycle , could be simple debris blockage, or vent valve not opening,play safe autoclaves are like accidents waiting for a place to happen -steam under pressure..
Posted on Feb 21, 2019
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
SOURCE: My autoclave ses 2000 turns
I would say that there is a problem with the thermostat. There is a small relay that pulls in to heat. And the relay opens when the thermostat reaches desired temp. I say it is the thermostat because it is pulling in the relay and heating. And if it is going to 107 degrees every time then the thermostat is pulling in the relay. So that tells you that the relay is working properly. So it would have to be the thermostat.
Posted on Sep 14, 2010
Tips for a great answer:
Mar 06, 2018 | Science & Laboratory
Jun 03, 2014 | VWR As12p Autoclave Accu Steam Sterilizer...
Apr 16, 2014 | ESCHMANN Little Sister Autoclave Tabletop
Jan 29, 2014 | ESCHMANN Little Sister Autoclave Tabletop
Apr 29, 2013 | Science & Laboratory
Feb 23, 2011 | Exercise & Fitness
Sep 06, 2010 | Health & Beauty
Sep 29, 2017 | Science & Laboratory
186 people viewed this question
Usually answered in minutes!
Step 2: Please assign your manual to a product: