Question about Zeiss Humphrey 740 Visual Field Analyzer
When we select a field the pointer on the touchscreen is down and to the right of where you are touching the screen
Posted by Anonymous on
You sjould press the patient button and the same time turn it the instrument on keep in button press until you see in tje display the calibration message of the touchscreen
Posted on Oct 30, 2014
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
Tips for a great answer:
Nov 13, 2016 | Janome Sewing Machines
Feb 23, 2016 | Dragon Touch Touch R10 10.1 Google Android...
Press the Windows logo key + X.
Click the little arrow next to Human Interface Devices to expand the list.
Click the touch screen driver (in my case, NextWindow Voltron Touch Screen).
Right-click, and select Disable from the list.
Click Yes on the dialog box that asks if you are sure you want to disable the touch screen driver.
When you want your touch screen to work again, follow the process above, EXCEPT when right-clicking the touch screen driver, select Enable
I’m happy to help further over the phone at https://www.6ya.com/expert/steve_52488d61fac93877
Sep 26, 2013 | ASUSTeK COMPUTER Asus VivoBook...
Apr 29, 2011 | Samsung SGH-A877 Impression Cell Phone
Jan 06, 2011 | GTCO CalComp Drawing Slate II
Jul 12, 2010 | Dell Vostro 1000 Notebook
Mar 19, 2010 | Samsung SGH-A877 Impression Cell Phone
Dec 28, 2009 | Nintendo DS Console
Nov 24, 2008 | Office Equipment & Supplies
320 people viewed this question
Usually answered in minutes!