Question about Zeiss Humphrey 740 Visual Field Analyzer
Aperture may be misaligned
I believe "state 19" is a weak main illumination bulb. Not sure if it's the same on your 740, but that's what the problem was with our 750i. Even though the bulb appears to be working, they dim over time and won't pass the start up calibration checks. On ours, there is a round access panel on top of the unit to service this bulb. Just be careful removing the bulb and don't bump any of the clear optical wheels. Part # 2660021106082 (for 750i. not sure about 740) Call Zeiss to confirm. Realize this is an old post so info is for future readers.
Posted on Jul 27, 2015
Save hours of searching online or wasting money on unnecessary repairs by talking to a 6YA Expert who can help you resolve this issue over the phone in a minute or two.
Best thing about this new service is that you are never placed on hold and get to talk to real repairmen in the US.
Here's a link to this great service
Posted on Jan 02, 2017
Tips for a great answer:
Jun 07, 2018 | Zeiss Science & Laboratory
Nov 04, 2014 | Zeiss Humphrey Acuitus 5015 Autorefractor...
Aug 04, 2014 | Zeiss Humphrey 740 Visual Field Analyzer
May 24, 2014 | Zeiss Humphrey 740 Visual Field Analyzer
Mar 21, 2014 | Zeiss Humphrey Acuitus 5015 Autorefractor...
Aug 26, 2013 | Zeiss Humphrey Acuitus 5015 Autorefractor...
Mar 29, 2013 | Zeiss Humphrey Atlas Eclipse 992 Corneal...
567 people viewed this question
Usually answered in minutes!